E1: aborted measurement – the measurement timeout is over, the measurement had to be aborted (the patient was moving).
E2: manually interrupted – the measurement was stopped by pressing a button
E3: battery rundown – the batteries exhausted during the measurement
E4: batteries changed – the batteries were replaced during the measurement
E5 (card(X)plore only): previous measurement was too close – minimum time restriction between two automatic BP measurements was not over.
E8: pressure limit exceeded – the pressure in the pneumatic system exceeded the preset pressure limit
E9: temporary disturbance – external electric signal (static discharge) disturbed the operation of the device
E31: cuff missing or loose – there was no cuff connected to the device or the cuff may be too loose on the patient’s arm
E32: cuff tubing clogged – the cuff is clogged or the rubber tube is broken
E33: cuff leaking or loose – there is a hole in the cuff or it is very loose on the patient’s arm
E34: cuff not on arm – the patient was not wearing the cuff
E40: plan is not running
E41: no flash card – there is no flash card int he device, flash card is insreted incorrectly or there is a communication error.
E42: incompatible card or no data – the inserted flash card is incompatible or does not contain measurement data
E43: different measurement plan – the flash card does not belong to the running plan
E44: faulty plan – measurement plan is damaged or contains incorrect parameters
E45: date-independent plan – time limit exceeded between creating and staring of a measurement plan
E90: device error – the device could not measure due to a hardware error
E99: device error – the device does not start automatic measurement (until the next programming)
Other error codes (apneABP only)